Portable Ultrasound Unit (Model ProM-660) Advance pain relief using sonic stimulation of soft tissue cells.
Three Treatment Modes (Low, Medium and High)
Easy-to-use Push Button Control
Packaged in attractive PVC case
Sold complete with AC Adapter, Conductive Gel, Manual, and carrying case
The ProM-660 generates deep heat within body tissues for the treatment of selected medical conditions such as relief of pain, muscle spasms, and joint contractrues, but not for the treatment of malignancies. Great for a second office unit, mobile therapy, sporting events, start up clinics, prescription to patients. Ultrasound is a form of deep heating in which sound waves penetrate into the soft tissues. It is especially useful in relieving acute pain and aids in tissue healing. This portable ultrasound unit is great for patients who need therapy away from a office or hospital setting. This unit is an FDA approved portable therapeutic ultrasound unit, now with 3 power output settings.
The precise parts are assembled and tested under strict process.
Designed with single chip micro processor, the quality is assured.
Precious alloy round headed probe creates smooth, nice-touching surface to the skin.
Attractive exterior fits to the human body and easy to hold and conveinient to use.
Micro-computer single button, user freindly.
On/Off button also functions to set output intensity.
Three output intensity selections to meet different requirements.
Frequency: 1 MHz
Pulse Width: 5.4 ms
Repetition Rate: 150 Hz
Temporal maximum power: 8W
Temporal maximum effective intensity: 1.13 W/cm2
Effective maximum effective intensity/Effective average effective intensity: 1.13/0.65 = 1.738
ERA: 7.06 square centimeter
Auto-time setting: 30 minutes
Output intensity: low, medium and high
Power: AC Adaptor (DC24V/300mA)
Size of main unit: 172mm(L) x 54mm(W) x 42mm(H)
Weight of main unit: 120gm
All values: +/- 10%
Ultrasound equipment generates high frequency sound waves that are transferred to a specific body area via a round headed probe. the sound waves travel deep into tissue and generate heat. As the probe glides over the skins surface, sound waves penetrate the skins surface causing soft tissues to creating deep heat. In turn, the heat induces vasodilatition: drawing blood into the target tissues. The generated deep heat is found to help relieve pain and reduce muscle spasms.
Ultrasound Therapy Device
Sun Conductive Gel - 2.8oz
DC plug adaptor
Functional Test Of Ultrasound Action
Place the probe horizontally then apply several drops of water on the surface of the probe. Turn the device on, you can observe the ultrasound action. The water drops on the probe start to perform one million vibrations per second with slight atomization phenomenon.
Low: Water drops vibrate softly
Med: The water drops vibrate stronger, and with atomization
Hi: The water drops vibrate very strongly, and with atomization phenomenon
Do not use over or near bone growth venters until bone growth is complete.
Do not use over a healing fracture.
Do not use over the eyes.
Do not use for patients with implanted neurostimulation systems because tissue damage can occur at the location of implanted electrodes resulting in severe injury or death. This can also damage the system components.
Do not us to treat malignancies nor in the region with malignant tumors.
Do not use for patients with demand type cardiac pacemakers.
Do not use on someone who is pregnant.
Do not use over ischemic tissues in patients with vascular disease where the blood supply would be unable to follow the increase in metabolic demand and tissue necrosis might result.
Do not use over the cartoid sinus nerves or arteries, laryngeal or pharyngeal muscles.
Do not use on patient with hemorrhagic diatheses.
Do not use over an area of the spinal cord following a laminectomy, i.e., when major covring tissue has been removed.
Do not use over anesthetic areas.
Avoid bony prominences.
When using ultrasound, keep the sound head moving while maintainging contact with skin.
If the treatment becaomes uncomfortable, inform or contact your physician.
Do not immerse the ultrasound device in water or any other solvent.
Do not use over metallic implants, especially prostheses with a cement-matrix.