Finally, a stronger, better, yet still portable ultrasound unit
is here! The Portable Ultrasound US Pro 2000 is a therapeutic
ultrasound device that generates deep heat through a piezoelectric
crystal sound head. Unlike electrical modalities, ultrasound is
unique in that the longitudinal waveform associated with sound
is not electromagnetic in nature. The piezoelectric crystal converts
electrical energy into a strong vibratory sound wave or ultrasound,
a sound beyond the normal range of our hearing.
Sound waves cause vibration in soft tissue cells, increasing
temperature and inducing vasodilation
Treatment duration is determined by the medical practitioner
and may range from 2-5 minutes depending on the conditions treated
and anatomic target site.
Because air is a very poor sound conductor ultrasound gels are
used to conduct the wave energy to the skin.
Ultrasound Gel - 2oz (118ml)
6 Month Manufacturers Warranty
Quick Start Guide
Connect power wire to wire socket.
Connect AC adaptor to power.
Apply sufficient gel to skin at target area.
Switch on the power. It power signal light is on, start to
use Portable Ultrasound.
Place the treatment head on the skin and move it slowly over
the target area in a circular motion or in linear strokes at
a rate of about 1 inch per second. DO NOT stop moving treatment
head while in use.
Use special care on bony prominence's to avoid discomfort.
The power automatically turns off in 10 minutes or, if desired,
the power can be turned off sooner by pushing the ON/OFF switch
After using the US-Pro 2000, clean skin with water to remove
Wipe excess gel off the treatment head with a damp cloth.
Disinfect treatment head with antiseptic wipe. Do not immerse
treatment head in water or solvent.
When power shuts off, leave unit off for 30 minutes before
After every treatment, disconnect AC adaptor from power
Do not impact the treatment head or drop treatment head
on the floor.
Turn on the device, the power indicator light is on. The default
modulate duty cycle is 30%, indicator light of “L”
level is on; When the default working time becomes 0, the indicator
light is off. Press "MODE" button, Modulate duty cycle
is adjustable (30%, 40%, 50%).
Power Source: Switch Adapter
Input: AC100~240V, 50/60 Hz
Output: DC 15V/1.2A
Frequency: 1.0 MHz ±10%
Modulation Wave Shape: 100Hz ± 10%
Temporal Maximum Power: 9.60W ± 20%
Effective radiating area: 4.0cm2 ± 20%
Temporal Maximum Effective Intensity: 2.4W/cm2 ± 20%
Max BNR: 5:1
Beam type: Collimated
Material of Applicator: Aluminium
Timer: 5, 10, and 15 minutes
Duty Factor: 30%
Output Power 2.88 ± 20%, for 4.0cm2
Duty Factor: 40%
Output Power 3.84 ± 20%, for 4.0cm2
Duty Factor: 50%
Output Power 4.80 ± 20%, for 4.0cm2
The US-Pro 2000 Portable Ultrasound should not be used for
the treatment of malignancies nor in the region with malignant
Do not use on people with poor arterial circulation.
Do not use over area with bleeding or infections.
Do not use over the eyes.
Do not use on someone who is pregnant.
Do no use for patients with demand type cardiac pacemakers.
Do not use over the carotid sinus nerves or arteries and laryngeal
or pharyngeal muscles.
DO’S and DON’TS
Do follow instructions of a licensed practitioner.
Do keep the treatment head clean.
lnform your clinician if you have any problems or questions regarding the ultrasound.
Do clean cables with water and mild soap. Do not use rubbing alcohol or any other solvent.
Do not damage your Portable Ultrasound US-Pro 2000 by bumping, dropping, or rough use.
Do not immerse Portable Ultrasound US-Pro 2000 in water or other solvents.
Do not pull or kink cables.
Do not store below 32°F (0°C) or over 122°F (50°C).
The Current Solutions US-Pro 2000 Portable Ultrasound Unit is a therapeutic pulsed ultrasound unit that enables chronic pain control. Its piezoelectric crystal sound head creates deep heat and sound waves that generate vibration in soft tissue, which helps induce vasodilation and further relieves pain. The US-Pro 2000 model features three duty, power and timer settings and is ideal for both home and professional use.